15.01.2020Blue Light irradiation test results
During the test the irradiance, weighted by the blue-light hazard, was not exceeded 0.58 mW/cm2 measured on distance of 25cm (10”) for Orchid/Orchid-F light and 0.27 mW/cm2 for Butterfly model. All measurement made on maximum power. Which are, in relation to maximum allowed limit, 5.8% for Orchid/Orchid-F and 2.5% for Butterfly model.
Report in PDF: ADMETEC_BLUE_LIGHT_TESTING
18.01.2020Admetec’s Butterfly Improvement
We’re happy to announce that our bestselling wireless headlight “Butterfly” has got major improvement. Starting from October 2019 all Butterfly headlights, batteries and chargers will be provided with new power connector preventing previously occurred problems. Also, the headlight itself will have renovated adapter plug, containing soft lock for smoother attaching. For previous models, Admetec will provide repair services during upcoming years.
22.01.2020IDEM SINGAPORE 2020
Admetec Solution Ltd. will take part in the international dental show IDEM 2020 in Singapore on the 24-26 April 2020.
We are proud to exhibit our newest products and to present ultimate solutions in dental optics.
Visit us at our booth – hall: 6F, stand 09.